Shenzhen boda jingke biotechnology co. LTD
address：5th floor,okwei building, 447 donghai avenue west, yantian district,shenzhen
The basic principle of multi-point magnetic agitator is to use the principle of magnetic field of like repelling and opposite attracting to push the magnetic agitator placed in the container to make circular movement, so as to achieve the purpose of stirring liquid.
Shenzhen boda jingke biotechnology co., ltd. is a high-tech company, committed to the development and application of advanced materials, the company's own product technology, research and development projects including nano new materials, widely used in industry, medical, scientific research laboratories and other fields; It is the mission of the company to promote the innovation and industrialization of nanotechnology and to advocate new energy conservation and environmental protection.
Multipoint magnetic agitator is a laboratory instrument for liquid mixing. It is mainly used for stirring or heating at the same time to stir liquid or solid-liquid mixtures with low viscosity. Its basic principle is to use the magnetic field of the same repel, opposites attract principle, using magnetic field to push the magnetic agitator placed in the container for circular movement, so as to achieve the purpose of stirring liquid. With the heating temperature control system, it can heat and control the sample temperature according to the specific experimental requirements, maintain the temperature conditions required by the experimental conditions, and ensure that the liquid mixture meets the experimental requirements
Matters needing attention in the use of multi-point magnetic agitator:
1. If the agitator beats or does not stir, please cut off the power to check whether the bottom of the beaker is flat and the position is correct. Meanwhile, please measure it.
2, heating time is generally not too long, intermittent use to extend life, do not heat when stirring.
3. Medium speed operation can work for 8 hours continuously, high speed operation can work for 4 hours continuously, prevent violent vibration when working.
4. Power consumption: the power socket shall be three-hole safety socket, which must be properly grounded.
5. The instrument should be kept clean and dry. Solution should not be allowed to flow into the machine to avoid damage.